Originally Posted by eryk4381
OK sorry it took so long to get the resistance measurement. I took out my probe from the install and checked it at room temp. I'm reading 11.3K ohms.
What is everyone else reading on their probes???..
Thanks again for the help.
Sounds about right. It's a 10K ohm resistor, so it should read 10K at 25°C (77°F). Your room temp is likely slightly cooler, hence the slightly higher resistance.
Did you try wiggling the wire at every point along it's length? If the probe wires got pinched or something, doing that would show a huge jump in the resistance. With the probe connected to controller, did you try wiggling the wires right at the contacts to see if it caused any changes in the readout?