
05062011, 11:18 PM

#1

Feedback Score: 0 reviews
Join Date: Mar 2011
Location: Las Vegas, NV
Posts: 900
Liked 76 Times on 58 Posts Likes Given: 219

New Refractometer FG Formula?


Has anyone used Sean Terrill's new refractometer formula for determining final gravities? He says that it is "on par" with the accuracy of using a hydrometer. I first learned about it on this Basic Brewing podcast episode, where he discusses the results of testing his new formula with the help of the podcast's audience. The details of the formula, along with an easy to use spreadsheet can be found here: http://seanterrill.com/2011/04/07/re...erfgresults/
Any statistics/math people looked into this? Has anyone done their own testing of his new formula?



05072011, 04:48 AM

#2

Feedback Score: 0 reviews
Join Date: Aug 2010
Location: McLean/Ogden, Virginia/Quebec
Posts: 7,754
Liked 1032 Times on 817 Posts Likes Given: 32

No. I've found too much dispersion in my own experience to consider use of a refractometer for FG estimation.
Thought it was funny that he called the linear more "accurate" but a bit less "precise" than the cubic. Here's a "new new cubic" which is, in his terminology, extremely accurate (mean error 0) and equally precise (standard deviation 0.0013) as the "new cubic": FG = 0.9993 – 0.0044993*RIi + 0.011774*RIf + 0.00027581*RIi² – 0.0012717*RIf² – 0.0000072800*RIi³ + 0.000063293*RIf³. IOW all you have to do is remove the bias for perfect "accuracy".
And removing the bias leads to a better fit  reduces the mse by bias^2.



07192011, 04:49 PM

#3

Feedback Score: 0 reviews
Join Date: May 2010
Location: Leadville, CO
Posts: 557
Liked 14 Times on 13 Posts Likes Given: 1

AJ: Could you expand on that? If by "bias" you're referring to the zeroorder coefficient, I set that to unity when doing the curve fits (not afterward) in order to hopefully improve the results that would be generated outside the interpolation ranges. If RIi = RIf = 0, then the FG result should be 1.000. You can certainly question the validity of that assumption, but I don't see how it can be compared to arbitrarily adjusting one of the coefficients after the fact.
From my perspective, the linear correlation *is* both more accurate (i.e. the mean deviation is smaller) and less precise (the standard deviation is larger). I'm entirely open to the possibility that you've noticed something I missed though.
Finally, what constitutes "too much dispersion" may be different for you than it is for others  especially given that most home brewers are apparently comfortable with hydrometers that can, at best, be read to ±0.001 SG.
Sean



07202011, 01:53 AM

#4

Feedback Score: 0 reviews
Join Date: Aug 2010
Location: McLean/Ogden, Virginia/Quebec
Posts: 7,754
Liked 1032 Times on 817 Posts Likes Given: 32

Quote:
Originally Posted by a10t2
AJ: Could you expand on that? If by "bias" you're referring to the zeroorder coefficient, I set that to unity when doing the curve fits (not afterward) in order to hopefully improve the results that would be generated outside the interpolation ranges. If RIi = RIf = 0, then the FG result should be 1.000. You can certainly question the validity of that assumption, but I don't see how it can be compared to arbitrarily adjusting one of the coefficients after the fact.

Had to go back and look at this  no idea what I was talking about at a first look. I'm pretty sure that what I had in mind is that a biased estimator (all the ones he mentions) have rms errors equal to the square root of the sum of the squares of the bias and the standard deviation squared. Thus if you have for the new cubic a mean error of 0.0007 and a standard deviation of 0.0013 the rmse will be sqrt(0.0007^2 + 0.0013^2) = 0.00148. If the estimator has 0.0007 subtracted from it (by knocking that amount off the constant term making it 0.9993 instead of 1.0000 then the mean error will be 0 and the rmse 0.0013.
Quote:
Originally Posted by a10t2
From my perspective, the linear correlation *is* both more accurate (i.e. the mean deviation is smaller) and less precise (the standard deviation is larger). I'm entirely open to the possibility that you've noticed something I missed though.

The linear estimate is the most accurate since its mean error is 0.0001 as opposed to larger values for the cubics but the cubics can be simply repaired to have 0 mean error as, of course, can the linear, by simply subtracting the mean error from the constant term. Thus they are all, with simple modification, can have the same accuracy (bias) and the rmse is totally determined by the standard deviations. As the "new cubic" and "new new cubic" have the same standard deviation they are the best with the new new being a better (in the mmse sense) estimator than the new because of the removal of the bias.
Quote:
Originally Posted by a10t2
Finally, what constitutes "too much dispersion" may be different for you than it is for others  especially given that most home brewers are apparently comfortable with hydrometers that can, at best, be read to ±0.001 SG.

I've seen discrepancies of 1 °P or more. That's enough to discourage me from using a refractometer. I can read a hydrometer to 0.1 °P and it's not that much more work (IMO).





