There is an offset in the auber PIDs to help calibrate the temp sensor. It is a +/- value you can set. Read the manual. The setting variable you need to change is "Pb".
If you are using a thermocouple, good luck. If the PID changes temp (i.e., gets warm), the reading will be incorrect. This does not apply to thermistors and RTDs.